|Inspection / Service
|Clean external surfaces, tray, and chamber with mild soap and distilled water. Wipe dry.
|Clean door gasket and mating surface with a damp cloth.
Inspect gasket for damage, replace gasket if necessary.
|Empty water from condensing tank. Clean tank with diluted bleach solution of 1/4 cup bleach : 1 gallon water and a brush.
Rinse tank, then fill to the minimum water level indicator line.
|Visually inspect components for damage that could cause problems during operation or unsafe operation.
|Drain / Refill Reservoir
|Using the drain hose, drain all water from the reservoir.
Refill reservoir with distilled water.
|All fasteners must be present and fastened securely.
|Inspect power cord and all wiring for damage.
|Be sure all electrical connections are tight.
|Remove and clean filter using a mild soap solution and a brush, Place in ultrasonic cleaner if necessary.
Rinse with distilled water.
|Clean condensing tank level sensors with mild soap solution.
|Date of Service: